- CFD-flow simulation in the measuring piping section
- under consideration of piping geometry and details as well as further installations like elbows, T-pieces and control valves at the inlet and outlet of the flowmeter
- Calculation of required length of piping sections
- Calculation of the pressure drop
VACOMASS® simulation presents results of a 3D-CFD (Computational Fluid Dynamics) simulation of the flow in the piping section. The detailed piping section, consisting of different parts like elbows, T-pieces, reduction and expansion but also control valve and flow meter sensor, is designed mechanically using a 3D-construction tool.
The complete geometry is read into a CFD-software and the defined body volume is divided into many small volumes (grid generation). Then border and initial conditions (like gas type, gas flow, pressure, temperature and other at the inlet and geometrical borders of the body) must be defined, the turbulence model needs to be selected and the iterative calculation can be started.
The flow in the piping section will get computed. Possible turbulences and flow stalls, which produce an unnecessary pressure drop and which must be paid later constantly by the operator, become visible and can be reduced or even avoided by an adaptation of the geometry of the piping section (e.g. increase of the length of the expansion piece), taking local situation into consideration.
Furthermore the computation can be done at different opening degrees of the control valve whether the simultaneous flow profile correction of the flow signal must be used or not.
Last but not least the pressure drop of the control valve as well as the pressure drop of the whole piping section can be calculated. So it can be examined if the requirements by the consultant/ customer respectively limitations given by existing blowers can be fulfilled or not.